Lin H.-L.,Hunan Province Environmental Monitoring Cerner |
Lin H.-L.,State Environmental Protection Key Laboratory of Monitoring for Heavy Metal Pollutants |
Gan J.,Hunan Province Environmental Monitoring Cerner |
Gan J.,State Environmental Protection Key Laboratory of Monitoring for Heavy Metal Pollutants |
And 9 more authors.
Guang Pu Xue Yu Guang Pu Fen Xi/Spectroscopy and Spectral Analysis | Year: 2015
A method for determination of beryllium in soils and sediments by microwave-acid digestion/graphite furnace atomic absorption (GFAA) is described. In this paper, the working conditions of the instrument are optimized, the drawing of calibration curve is expounded, the pretreatment process of soil and sediments (including microwave heating process and the selection of digestion system) is discussed, and the interference of coexisting elements is examined. The sample was pretreated by microwave digestion parameters using HNO3/HCl/HF mixed acid system. The method is fast and simple without matrix modifier, and has no interference by coexisting ions, and has high repeatability and reproducibility. Under the optimal experimental conditions, the limit of detection (LOD) is 0.004 9 mg·kg-1 (sample quantity 0.200 0 g, sample volume 25 mL), and the limits of quantitation (LOQ) is 0.020 mg·kg-1. This method is used to measure the standard samples and actual samples, whether in the laboratory, or between laboratories, has good accuracy and precision. © 2015, Science Press. All right reserved. Source