Hua F.,Beijing Institute of Microbiology and Epidemiology |
Hua F.,Beijing Key Laboratory of POCT for Bioemergency and Clinic |
Hua F.,Taishan Medical University |
Hua F.,Clinical Diagnostic Center |
And 20 more authors.
Scientific Reports | Year: 2015
Francisella tularensis is a potential biowarfare/bioterrorism agent and zoonotic pathogen that causes tularemia; thus, surveillance of F. tularensis and first-level emergency response using point-of-care testing (POCT) are essential. The UPT-LF POCT assay was established to quantitatively detect F. tularensis within 15 min, and the sensitivity of the assay was 104CFU · mL-1 (100 CFU/test). The linear quantitative range covered five orders of magnitude, and the coefficients of variation were less than 10%. Except Shigella dysenteriae, UPT-LF showed excellent specificity to four strains that are also potential biowarfare/bioterrorism agents and 13 food-borne pathogenic strains. Samples with pH 2-13, high ion strengths (≥ 2 mol · L 1 solution of KCl and NaCl), high viscosities (≤ 50 mg · mL 1 PEG20000 or ≥ 20% glycerol), and high concentrations of biomacromolecules (≥ 400 mg · mL 1 bovine serum albumin or ≥ 80 mg · mL 1 casein) showed little influence on the assay. For practical utilization, the tolerance limits for seven powders and eight viscera were determined, and operation errors of liquid measurement demonstrated a minor influence on the strip. Ftu-UPT-LF is a candidate POCT method because of its excellent sensitivity, specificity, and stability in complex samples, as well as low operation error.
Zhao Y.,Beijing Institute of Microbiology and Epidemiology |
Zhao Y.,Beijing Key Laboratory of POCT for Bioemergency and Clinic |
Wang H.,Beijing Institute of Microbiology and Epidemiology |
Wang H.,Massey University |
And 14 more authors.
Scientific Reports | Year: 2016
The rapid high-throughput detection of foodborne pathogens is essential in controlling food safety. In this study, a 10-channel up-converting phosphor technology-based lateral flow (TC-UPT-LF) assay was established for the rapid and simultaneous detection of 10 epidemic foodborne pathogens. Ten different single-target UPT-LF strips were developed and integrated into one TC-UPT-LF disc with optimization. Without enrichment the TC-UPT-LF assay had a detection sensitivity of 10 4 CFU mL â '1 or 10 5 CFU mL â '1 for each pathogen, and after sample enrichment it was 10 CFU/0.6 mg. The assay also showed good linearity, allowing quantitative detection, with a linear fitting coefficient of determination (R 2) of 0.916-0.998. The 10 detection channels did not cross-react, so multiple targets could be specifically detected. When 279 real food samples were tested, the assay was highly consistent (100%) with culture-based methods. The results for 110 food samples artificially contaminated with single or multiple targets showed a high detection rate (≥80%) for most target bacteria. Overall, the TC-UPT-LF assay allows the rapid, quantitative, and simultaneous detection of 10 kinds of foodborne pathogens within 20 min, and is especially suitable for the rapid detection and surveillance of foodborne pathogens in food and water.